Customer Care 877.393.8676 904.280.1900
More information 855.655.2020 or email
What Does Wave Design?
Wave software recommends a starting custom lens design regardless of whether it's a single vision, front surface multifocal (toric or bi-toric), ortho k, toric-ortho k , toric, bi-toric lens or free form asymmetric lens. Designs have multiple aspheric curves as small as 10 microns wide depending upon the shape of the cornea. Wave automatically adjusts the front surface optics to work with the surface topography to create a higher performance lens than common off the shelf lenses.
Wave first imports raw topography data to convert to a corneal model to work with. Next a starting lens design is suggested. All Wave designs have multiple aspheric curves as small as 10 microns in diameter to closely follow the corneal shape.
Default designs tend to be larger as Wave uses the entire corneal map to 'know' the corneal periphery. Great emphasis is given to closely duplicating the corneal periphery into the lens periphery, resulting in better centering. At any time the designer may increase or decrease OAD (8.5-15mm), center or edge thickness.
The designer has full capability to alter the fit as they like. Wave designers think in term of tear film thickness when making adjustment to fit. Looking at the simulated fluorescein map, the designer 'grabs' and moves one of four control balls to increase or decrease tear thickness across the lens. Individual or combined control of up to eight meridians affects tear film in 90, 180 or 360 degree segments of the lens. The same control balls are used to adjust OZ, IZ and edge diameter and bevel lift.
Design changes with an easy drag-and-drop approach are seen in the simulated fluorescein map. Many other design changes are possible in addition to compound over-refractions, if necessary, to refine visual acuity. And Wave's automatic cross cylinder feature compares corneal cylinder to refractive cylinder with an option to incorporate the expected residual internal astigmatism on the front surface of the first lens.
Scroll and click on highlighted areas.
Learn more, click for a video.
Rx $ Multifocal Input
Tear Film Graph
Optic Zone Control
Opens Settings Window (see below). Design your own templates. Select for quick design.
Not able to get a good topographic map? Insert trial lens data and then use Wave to design.
Want to start over? Revert to default starting design
After making multiple adjustments to a design, have Wave recalculate to better sync adjustments.
Import a previous design to compare to your current design. Graphs show both sets of data at once. Use older design to help guide adjustments.
Import a design from an outside source.
Auto design a semi-scleral lens with an alignment fit. Make adjustments as necessary.
Auto design a semi-scleral lens that vaults the cornea. Make adjustments as necessary.
Enter a (drop down) value for any of these parameters. Save and name your "definition" as a template.
Wave includes several common definitions. You may add as many as you wish. When fitting a patient with this feature, after selecting a definition you may change any of the parameters for that patient's needs.
Additional design parameters that are part of a definition.
Capabilities & Specifications
Lens Design: Rotationally symmetric, geometrically symmetric or asymmetric front and back surfaces with atoric spherical effect optics
Materials: Most common GP materials
Optical Design: Single Vision, Multifocal (center distance, center near or segmented), Aspheric Toric, Aspheric Bitoric with or without front surface prism, Myopic and Hyperopic Ortho keratology
Compound Optics: Combine multiple elements (multifocal, toric, bitoric, and reverse geometry) into one design, even a spherical center with a toric periphery!
Power Range: Sphere +/- 30 diopters, Cylinder +/- 30 diopters, Add +0.25 - 5 diopters
Dimensions/thickness Control: Diameter 8.5 -15mm, center thickness 0.12-1.0mm, edge
thickness 0.10 - 0.30mm; Soft - Diameter 13 - 15mm, center thickness 0.08 -1.0mm, edge
thickness 0.10 - 0.30mm
Design Controls: Initial Rx, over refraction, compound over refraction, diameters of optical zone, intermediate zone, edge, bevel
Tear Film / Zone Diameter Control: Rotational designs: drag and drop global control of zone diameters and tear film clearance of optical zone, intermediate zone, bevel and edge, 360 degrees around the lens. Asymmetric and Geometrically symmetrical designs: drag and drop control of zone diameters and tear film clearance of optical zone, intermediate zone, bevel and edge at 8 individual meridians controlling a 90,180 or 360 degree segment of the lens.
Software Platform: Windows 98SE or better
Data Source: Scout or Piccolo Topographer, Medmont Topographer, Oculus Topographer.
Copyright © 2007-2015 Wave LLC. All Rights Reserved.
228 Canal Blvd Suite 3
Ponte Vedra FL 32082 USA